Wherever they end up!
Track & Confirm
and Use information
click to return to Titer
Garage product sheet
This is a Chillable holder used to cool skirted and unskirted microtiter
plates. This device can be used with various automated liquid handlers and
benchtop laboratory use. It will keep the contents chilled for more than
~1 hr* at room temperature. 12 mini Gel paks included
Define the TITER GARAGE using the respective software utilities for your
To do this increase the definition height Z + 25.15 mm.
Plate use Biomek 2000.
Place the plate in a standard labware holder measure from the top of the
plate to the stainless steel deck, and record this value(Platetop). Then
place the same plate in the Titer Garage and measure the same distance.
(Titer Garage top) subtract the (Titer Garage top) – (Platetop) = new Z
offset for the device holder. Copy a device ‘labware holder ‘ name it
‘Garageholder’. Add the determined Z offset in the Labware position
deference section. Save. Use this definition under this type of plate
additional definitions can be created for other plates as needed.
Device Height: ……….. 29.30 mm
Plate Height………….. 25.15 mm
Positions as a titer format plate at the base.
Place as many mini gel paks into the TITER GARAGE locate a representative
plate and refrigerate for 24 hours to give the gel paks shape (be careful
not to over stuff, the paks may swell when frozen). The Beckman foil seals
or packing tape work great to seal the paks in place . This device can
also be used as a collar holder for the DynamifoldÔ
. If you encounter any difficulty with this product feel free to contact
Jeff Cahlik at Acme-Automation.
*Depends on a number of conditions
like plate type,well depth, well volume